Magnetic Induction Process
Magnetic induction test method according to the standards:
ISO 2178:2016-3, ASTM 7091
Contact method: The excitation current generates a low-frequency magnetic field with a strength that corresponds to the distance between the probe and the base material. A measurement coil measures the magnetic field. In the instrument, the obtained measurement signal is converted into the coating thickness values via the characteristic probe output function, i.e., the functional correlation between the probe signal and the coating thickness.
Main fields of application: Non-magnetizable coating materials on magnetizable base material.
- Electroplated coatings of chrome, zinc, copper or aluminum on steel or iron
- Paint, enamel, lacquer or plastic coatings on steel or iron
Your contact to FISCHER
HELMUT FISCHER GMBH
INSTITUT FÜR ELEKTRONIK UND MESSTECHNIK
Helmut Fischer GmbH
Industriestraße 21, Sindelfingen/Deutschland
Phone: +49 (0) 70 31 / 3 03 - 525